Data is not reaching the computer on any channel and the photogate status check does not work or Data Monitor shows only noise.
The 6500 cable may be attached to the wrong port on the interface board. It should be plugged into the port furthest from the keyboard: the "back" port.
Creation Date: 01/1/2000
Last Modified: 07/21/2003