Sine Wave Generator output is not measurable.
The WA-9876 Sine Wave Generator uses a Class D amplifier at the output. For this type of amplifier, neither side of the output is referenced to Earth ground. If you connect a scope directly to the output, then you will short out the signal.
Creation Date: 07/14/2009
Last Modified: 12/3/2013